Caption: APL calibration but the bath computer was not working. We took points by hand and computed the calibration.
The three calibrations (cal1,cal2,cal3) are compared. The data was reprocessed for the three different calibration coefs. The means over the entire cruise were cal1-cal2=0.025+/-.03 and cal3-cal2=0.009+/-.04. Cal2 was used for the real time data. Uncertainty from calibrations is about 0.02 with excursions to 0.15.v